Light-cured rubber-dam liquid
Light-cured RUBBER-DAM liquid is dedicated for:
- gingiva protection during dental treatments: whitening, etching, microabrasion, sandblasting etc.
- sealing rubberdam.
Suitable for occlusal stamp technique.
urethane resins, inorganic fillers.
- Syringe containing 1.2 ml of the preparation + disposable applicators
– MEGA PACK – 4× Syringe containing 1.2 ml of the preparation + disposable applicators
- new quality of work
- perfect protection of soft tissues
- wide range of use
Method of use:
For gingiva protection during dental treatments:
Clean and dry gingiva. With the use of appropriate instruments (e.g. buccal and labial retractor) immobilize lips and a tongue in order to avoid contact with the liquid. Remove the syringe lock and install the included disposable applicator. Apply RUBBER-DAM LIQUID along the application area 4-6 mm wide and approximately 1.5-2 mm thick, at the same time overseeing the amount of the preparation being applied (Pic.1). If possible, squeeze RUBBER-DAM LIQUID through interdental space. Cure the preparation for 20-30 seconds using polymerization lamp (Pic.2). Conduct the curing process evenly in a scanning motion. Then, proceed the dental treatment. After the treatment is completed, lift off polymerized RUBBER-DAM LIQUID using the instrument on the either side and remove.
In order to seal rubber dam:
Isolate the working area (tooth, teeth, alveolus) using RUBBER-DAM. In order to seal rubber dam, apply the layer of the liquid on the edge of the gingiva and the rubber dam (Pic.3, Pic.5). Cure the preparation for 20-30 seconds using a polymerization lamp. Conduct the curing process evenly in a scanning motion (Pic.4, Pic.6). Then, proceed the dental treatment. After the treatment is completed, lift off polymerized RUBBER-DAM LIQUID using the instrument on either side and remove.
Immediately after the use withdraw the syringe plunger and place the original cap on the container to avoid polymerization of the product in the syringe. The product is intended for reusing, while the applicator is designed for disposable use. Reusing the applicator may cause the hazard of re-infection.
FOR PROFESSIONAL DENTAL USE ONLY